Used in our labs to ensure the integrity of all diamonds that pass through our facilities.
Loose polished stones can be tested in the 0.05 - 10cts size range. The sample is placed table down on the fibre optic probe and the test button is pressed.
All synthetics, simulants and around 2 % of natural diamonds (TypeII and IaB) will refer for further tests. These stones can then be tested on the DiamondView.
For stones in mounted jewellery, the sample dish can be removed to release the fibre optic probe,
and can be tested using the ‘Auto-detect’ feature. Around 98% of stones will pass and will require no further testing.
DiamondSure is approximately 17cm in width, 26cm in diameter, 10cm height.
The latest version of the instrument comes with a detachable probe. As long as the probe is in good contact with the stone alone, and not any surrounding metal, then a measurement can be made.
This message appears if the calibration arm is not in place during start up or the factory calibration data doesn’t match the field data. Ensure the calibration arm is in place and press left hand button again.
The DiamondSure makes a measurement of the light returned from the diamond under test and can detect the presence of strong overhead lighting. Either move the DiamondSure away from direct overhead lighting or move the light source if possible.
No. Replacement lamps must be purchased from the instrument manufacturer.
The instrument performance may not necessarily be impaired. The unit should only be returned for servicing if the referral rate for natural diamonds is noticeably high or the unit fails to initialise.
Check the fibre tip is not below the white plastic disc (if it is then adjust height). Replace lamp and re-align. Replace fibre. Return to manufacturer if previous actions do not remedy the problem.
The detachable probe can be used to measure stones mounted in jewellery. For stones up to 0.3 carats an insert is not generally needed. The smallest insert is suitable for stones up to 0.5 carats, the next size insert is suitable for stones up to 1 carat, the next size insert is suitable for stones up to 2 carat and the largest insert is suitable for stones up to 4 carats.
For Support on IIDGR instrument issues follow the link below.